Battery Two-zone Thermal Shock Chamber
Battery two-zone thermal shock chamber is used to test the bearing extent of the material structures and composite material in an instant and continuous high temperature and extremely low temperature environment, which is in the shortest time to test its thermal expansion and contraction caused by chemical change or physical harm.
|Packaging||Each set protected by resin fiber and PP film, then put into Strong wooden case with operation manual and vedio inside.|
Battery Two-zone Thermal Shock Chamber
Battery two-zone thermal shock chamber is used to test the bearing extent of the material structures and composite material in an instant and continuous high temperature and extremely low temperature environment, which is in the shortest time to test its thermal expansion and contraction caused by chemical change or physical harm.The equipment is widely used in metal, plastic, rubber, electronics and other materials and it can be used as an important reference for improving product quality. It meets the standard of MTL, IEC, JIS, GJB…
- Perfect appearance and easy to operate.
- Japan LCD English microcomputer temperature controller, with high memory capacity, can save 100 programs, max 9999 cycles, each cycle max time 999hrs.
- Three chamber structures, high temperature chamber, low temperature chamber and test chamber, control automatically, stay and switch time adjustable.
- Perfect protection alarm functions: short-circuits, Leakage, over temperature of work chamber; compressor over pressure, over load, short water…
- Control interface will display alarm reason and provide check methods, machine has emergency stop button.
- Cold and hot shock temperature recover within 5 minutes, which complies with MTL, IEC, JIS, GJB and other international standards.
- With RS-232C communication port for connecting PC to control, record and save test report.
|Test method||Pneumatic operated doorswitch, two chambers|
|Basket internal dimension||320Wx230Dx250Hmm (20L)|
|External chamber dimension||1050Wx1825Dx1440Hmm|
|High temperature range||Room temperature ～+175℃|
|Low temperature range||Room temperature ～-50℃|
|Testing time||Maximum 1000 cycles, 30 minutes for each one|
|Defrost||Do defrost in every 20 cycles|
|Lifting switch time||≤10 seconds|
|Test hole for cable routing||Diameter 50mm on the left of chamber attach a silicone plug|
|External chamber material||High quality electrolytic steel plate spray with powder, antirust|
|Internal material||SUS#304 stainless steel|
|Thermal insulation material||High quality superfine glass fiber|
|Refrigerant||R404A / R23|
|Cooling method||Water cooling|
|Power supply||Three phase, 380V, 50Hz|
High temperature chamber:
Heater: NI-CR alloy electric heater
Draught fan: Share centrifugal fan when high temperature and do environmental temperature, preheating with axial flow fan.
Low temperature chamber:
Cooler: Fin cooler, regenerator
Draught fan: centrifugal fan
Pneumatic cylinder: Drive the air door of high, environmental, low temperature
Air compressor: Provide the compressed air for driving pneumatic door (supplied by buyer)
Cooling method: mechanical compression refrigeration
Refrigeration compressor: Semi-closed piston
Condensation: Stainless steel welded plate heat exchanger
Operate interface: 6" color LCD touch screen
Program memory capacity: 100 groups (modify by operator)
Setting range: Time: 1 minute ~ 99 hour 59 minute, cycles: 1~ 999 cycles
Resolution: temperature: 0.01 ℃; time: 1 minute
Input: T thermocouple
Control method: P.I.D control
Timer, over temperature and power failure protection, alarm, record test curve, emergency stop, display test time
Circulated cooling water: pressure: 0.2～0.4MPa, temperature: ≤30℃
Standard configuration: indication lamp of test chamber, time meter, test hole for wire routing, 6pcs castor, 4pcs adjust wheel.
Leakage and breaker, over temperature, low temperature protect device, exhaust valves, sample power control terminal, over temperature protection of high and low chamber, compressor over pressure, over heat protection, out of water relay, heat relay of fan, compressed air pressure switch, fuse, external alarm terminal